AGILENT 34401A LABVIEW DRIVER
Double check the com settings on the instrument and verify they match. From the block diagram you can see that the instrument is first initialized with the Serial Configuration and VISA resource name specified on the front panel. Many instrument drivers have a VI Tree VI you can use to view the entire instrument driver hierarchy. Message 3 of 4. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
|Date Added:||22 December 2005|
|File Size:||56.31 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
You also can use the Instrument Driver Finder to open example programs covering everything from analysis and presentation to using your instrument driver.
Message 3 of 4. Message 1 of 4. I’m not sure about instructions, but if you installed the driver and opened the library, you will get sub-vi’s which have a particular application purpose. Works like a charm. LabVIEW instrument drivers simplify instrument control and reduce test program development time by eliminating the need for you to learn the complex, low-level programming commands for each instrument.
I hope not to offend NI people http: For other instrument driver types, the examples will be available from the Application Examples subpalette from the corresponding instrument driver. Examples include VIs for reading a measured value or waveform and VIs for downloading waveforms or digital patterns to a source instrument. Feel free to write us at les.
Most Active Software Boards: Message 3 of 7. Then agileent Configure Measurement VI is called to set the instrument up for the desired measurement. I opened that one. I download directly from Agilent’s website.
How to Use an Instrument Driver in LabVIEW Tutorial
Double check the com settings on the instrument and verify they match. Have a great day!
Numerous Configuration VIs can exist, depending on the particular instrument. When all of these settings are set and the VI is executed the results will be displayed in the Measurements array indicator. The Initialize VI- the first instrument driver VI you call, establishes communication with the instrument. Has a 3 Application Examples and a “Getting Started” one which should get you on your way.
Reading Max/Min/Avg with labview from agilent a – Discussion Forums – National Instruments
I used these settings to run the program, so I think the baud rate and other stuff is set correctly. I just downloaded the file again and 34401aa the steps I described earlier. Message 4 of 7. Les Hammer Complete Test. If these suggestions do not solve your problem, please post a screenshot of the error so I can look more into the matter.
Concerning the proper configuration of the instrument, wgilent may be available in your user manual or from th e manufacturer. Generally, you need to call the Close VI only lbaview at the end of an application or when you finish communication with an instrument.
It would be extremely useful to show how to modify portions of instrument drivers for other use. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
I am having a similar agilen. Once the measurement is complete, the Close VI is called which performs an instrument error query and terminates the software connection to the instrument.
I opened the code to try to trouble shoot. I downloaded the most recent drive, it’s dated for Each VI corresponds to a programmatic operation, such as configuring, reading from, writing to, and triggering an instrument.
The current version is 4.